Hanna Iron Checker HI721.
- Easier to use and more accurate than chemical test kits
±0.04 ppm ±2% of reading accuracy
0.01 ppm (mg/L) resolution (500 points)
Large, easy to read digits
Auto shut off
- Dedicated to a single parameter
Designed to work with HANNA’s powder reagents
Uses 10 mL glass cuvettes
- Small size, big convenience
Weighing a mere 64 g (2.25 oz.), the Checker®HC easily
fits into the palm of your hand or pocket
Use for quick and accurate on the spot analysis
One button operation: zero and measure
Operated by a single AAA battery
- Ideal for:
Industrial, ground and treated waters, mining leachate monitoring and
agricultural irrigation water
- For use in fresh and saltwater
The HANNA HI 721 Checker®HC bridges the gap between simple chemical test kits and professional instrumentation. Chemical test kits are not very accurate and only give 5 to 10 points resolution while professional instrumentation can cost hundreds of dollars and can be time consuming to calibrate and maintain. The HANNA HI 721 Checker®HC is accurate and affordable with immediate results.
The new HI 721 Checker®HC portable handheld colorimeter features a resolution of 0.01 ppm (500 points) and ±0.04 ppm ±2% of reading accuracy. The HI 721 Checker®HC uses an adaptation of Standard Method 315 B.
The contoured style of this Checker®HC fits in your palm and pocket perfectly and the large LCD is easy to read. The auto shut-off feature assures the battery life will not be drained if you forget to turn it off.
The HI 721 Checker®HC is extremely simple to use. First, zero the instrument with your water sample. Next, add the reagent. Last, place the vial into the HI 721 Checker®HC, press the button and read the results. It’s that easy.
HI 721 Checker®HC is supplied with sample cuvettes with caps (2), powder reagents for iron (6), battery and instructions.
|Range||0.00 to 5.00 ppm (mg/L)|
|Resolution||0.01 ppm (mg/L)|
|Accuracy @ 25°C||±0.04 ppm ±2% of reading|
|Light Source||LED @ 525 nm|
|Light Detector||silicon photocell|
|Environment||0 to 50°C (32 to 122°F); RH max 95% non-condensing|
|Battery Type||(1) 1.5V AAA|
|Auto-off||after three minutes of non-use and ten seconds after reading|
|Dimensions||81.5 x 61 x 37.5 mm (3.2 x 2.4 x 1.5”)|
|Weight||64 g (2.25 oz.)|
|Method||adaptation of the EPA Phenantroline method 315 B, for natural and treated waters.|
Questions on Hanna Iron Checker HI721
No questions asked yet
Ask Your Own Question
Handy Little Kit Review by Ithacajosh
What a great kit! I test water daily, and I've used several iron test kits. Each one had a flaw, for one reason or another. Not this one. It's self-contained, easy to use, and accurate. If you follow the directions you cannot screw it up! (Posted on 1/22/2015)
Write Your Own Review
- Handy Little Kit Review by Ithacajosh